ICU Medical Bridge Ceiling Pendant (HFP-S/S+E)
1.ICU Bridge Type Ceiling-Mounted system,specially designed for intensive care rooms, the entire proocess wards, the recovery room.Supply flexible, first-rank nursing satisty patient's separate treatment and nursing demand.Unique design lies in it all hides the electrical cable and pipeline in the ceiling-mounted arm.
2.The modular design and multiple configurations.
3.As clients' requirements,providing reasonable ICU Combination Ceiling-Mounted Pendant(Apart Dry-Wet).
4.Advanced spray technology, the intelligent combination of innovative ways to make the bridge pendant with diverse functions.
5 Wet and dry functions can be combined with the freedom of columns, boxes and rotate arm,maximize access to the patient, rescue time to survive.
|Main Technical Data|
|Type||HFP-S+E ICU Bridge Type Celling Pendant(Apart Dry-Wet)|
|Opreating force (linear and horizontal) (N)||<50|
|Brake power (linear and horizontal) (N)||<100|
|Ceiling mount||By two ceiling frames at raw ceiling|
|Ambient Light (W)||2×54|
|Reading/Vistors'Lights||220V,2×15W LED spotlights|
|4 Gas socket(German Standard)||O2×2,VAC×1,AIR×1|
|Cable pipe clamp||4|
|1 Column lengths(mm)||700|
|3 Gas socket(German Standard)||O2×1,Vac×1,Air×1|
|Medical Gas Outlets Standards:||British, German, French, Metric, Ohmeda, DISS, Chemetron, Purit-Bennett standards optional|
|Drawer Layer|| 1/2|
|Optional Power Driven||USD200.00 additional|
|Electrical Sockets||can be customization as clients' request|
|Delivery time||7 days|
Δ Whole horizontal rail movement.
Δ This kind gas outlets, power sockets, network interfaces, lights placed on the bridge rail.
Δ Dry side and wet side for multifunction, apart wet and dry area, or together with wet and dry area is optional.
Δ Very simple, easy to work.
Δ Ceiling shelf flexible diversity.
Δ Enrichment equipment accessories make the work regional more orderly.
Δ All-round patient access.
Δ A totally closed design helps prevent dust accumulation and facilitates cleaning.
Δ All material adopt high-strength aluminium alloy extrusion molding, a surface
Δ Oxide anti-rust treatment.
Basic Configuration and Technical Parameter
1. Max capacity of function case(dry section): 150kg
Max capacity of function pillar(wet section):150kg
2. Effective shelving size of dry section tray: 690x400mm
Effective shelving size of wet section tray: 590x400mm
3. numbers of gas, electricity terminal of function case: 16 pieces
numbers of gas electricity terminal of function pillar: 40 pieces
4. Standard configuration of the product:
Gas socket---2x O2,1x Air,2x Vac
Power socket---power socketx12 pieces
Communication socket---RJ45x1, specification: classification 5, chip 8(or classification 6, chip 8)
1. Electrical Requirements Instructions:
--The electric power for each beds: 2.0 KW( Just for reference)
--In principle, Each beds should configure a separate power supply circuit/
2. Gas pipeline / power supply cable / communication cable shall be responsible for the relevant contractor
--Laying medical gas supply line to the point of hanging tower supply
Gas pipeline wear into the hanging tube (tower crane), the bending radius: R> = 30mm
Diameter requires oxygen / air: diameter 8mm, vacuum absorption: 10mm in diameter
--For laying power lines to locate the point of hanging tower
Length should be set aside to ensure that after the commencement of its ground clearance <1m
For power line:VV-3*2.5mm2, BV-1*6mm2
--Laying call/Monitor the communication line to the point of hanging tower positioning
When Nurse Call and monitor communications with the line, its length should be set aside to ensure
After expand it can be directly connected to the nurse call and monitor terminal
3. Additional openings on the device description
--Hole size provided by the nurse call nurse call vendor
--Hole size provided by the monitor terminal monitor supplier